The impact of work stress on Coriolis mass flowmeter
Published on：2015-3-30 11:02:22 Hits：
Abstract: Coriolis mass flowmeter has a direct measure due to the advantages of mass flow in China from the early 1990s into the industrial measurement field since more and more widely. Early mass flowmeter manufacturers claim that their products are not affected by the density, temperature, pressure, viscosity and other changes in measured media, with a large number of applications in recent years, found that the physical parameters of the measuring medium changes its measurement accuracy is influential. This paper discusses the impact of changes in measured media pressure on flow measurement accuracy.
Keywords: Coriolis mass flowmeter, measurement accuracy, measurement of media, working pressure, error
First, the Coriolis mass flowmeter works
Coriolis mass flowmeter is the use of fluid mass flow rate of the modulation effect vibrating tube oscillation phenomenon is the principle that the Coriolis force to mass flow measurement of mass flow purposes. Generally by the sensor and transmitter components.
- The various types of flowmet...[15-11-10]
- How to use glass rotameter[15-11-10]
- flowmeter selection principl...[15-05-05]
- volumetric flowmeter selecti...[15-04-28]
- Power Plant Coal Handling Sy...[15-04-16]
Sales Hotline: +86 519 400 888 7116
Technical line: (+86 ) 138 6108 2011
Fax: +86 519 86909383